TSP Sampling Procedures
For TSP sampling, the
following changes must be made to the filter holder:
Remove the impactor(s) from the preseparator assembly and multiple impactor adapter; remove the adapter.
Clean the cap and preseparator assembly every 5 sampling periods, or more frequently if soiling is observed.
After the sampler has been assembled, calibrated, verified to be in proper working order, and a filter loaded in the filter holder assembly, set flow rate adjustment in accordance with procedures as shown in the Operation Manual.
Follow all other procedures in the Operation Manual, noting that TSP is being sampled.
to Basic Operation Page